To ensure that resistance measurements remain accurate, we suggest yearly calibration. Certain components within CableEye Models M3U, M3UH, M4, HVX, and HVX-21 are analog components whose characteristics change slightly over time. This aging is a natural phenomenon common to all electronic components and may cause resistance measurements to gradually deviate from normal.
Companies interested in quality product and maximum yield will ensure their testers are calibrated. Those with quality certifications such as Six Sigma or ISO9000 will be required to calibrate.
A tester that is out of calibration will give inaccurate results. Some (maybe all) resistance measurements will be incorrect. Consequently, it may Pass what should Fail and, conversely, Fail what should Pass. In other words, product yield will be contaminated due to unreliable quality.
As a service to our customers, one month before your equipment reaches its recalibration anniversary, we will automatically send a letter to the person whose name is on record as the contact individual at your company advising this fact and the cost and procedure for recalibration.
You will need to send the tester (with power module if it's a low voltage tester) to CAMI Research or to an authorized international distributor for recalibration.
Recalibration can usually be completed within 1-2 business days of receiving your equipment. The time of round-trip transportation will, of course, depend on the shipping method you choose.
The equipment will be returned with a new calibration sticker and a CD. The CD contains the calibration data file and, if you also renew your warranty, the latest software release. Replace your existing calibration file with the new one. Customers will find installation instructions at Customer Support.
Note 3: Effective January 1, 2018, calibration services for serial port models M3 and M3A will require that the tester be upgraded to USB i.e. to an M3U.
Yes, if you purchase our optional Tester Calibration Verification board viz. item 780 for M3U testers, and item 780H for M3UH, M4, and the HVX series.
If your tester is fitted with the 4-Wire Kelvin Measurement Add-On option, you will also need the 4-Wire Calibration Verification Board (item 782).
CAMI Research has some of the best customer support and response times of any of our vendors, and I want you to know that this is truly and sincerely appreciated.
Hear more from our customers ...
Yes, if you purchase a Calibration Kit/s.
The Low Voltage (LV) kit is required for every tester (except the M2U series for which calibration is not applicable). It includes a calibration board with precision reference resistors, special software, a procedure, and cables. In addition, you will need your own benchtop Digital Multimeter that resolves to 0.1mV and is calibrated to NIST standards. If your tester is equipped with a capacitance measuring capability, then you will also need a Precision Capacitance Decade Box that is calibrated to NIST standards. We recommend Item 717C or an equivalent device.
To calibrate HVX-series testers, order a LV kit together with an appropriate High Voltage (HV) kit. HV kits include further software, a procedure, and special voltage-divider switch box/es.
Calibrating CableEye does not involve adjusting trimpots or any other mechanical adjustments. Instead, special software makes precise measurements under known conditions and computes a set of "calibration constants" within a defined tolerance for each test point. The resulting calibration file is generated at the end of the procedure. You then place this file, ending in ".cal", in your /Program Files/CableEye/Software folder. When the CableEye application starts, it looks for this file and uses its data to compensate for slight variations in the semiconductors.
In the unlikely event that the calibration software computes constants that lie outside of the predefined tolerance, the software stops, fails the calibration, and notifies the technician, asking whether s/he wishes to retry. If the calibration fails again, the equipment will need to be repaired by CAMI personnel.
In general, if your company has two or more CableEye testers and intending to calibrate yearly, it may be worth the cost and technician time to purchase the calibration kits. However, for HVX series testers in particular, we highly recommend you return your tester to us for factory calibration once every couple of year in order to benefit from the general inspection and cleaning described in Note 2 above.
Note 4: In addition, you will need your own benchtop Digital Multimeter that resolves to 0.1mV and is calibrated to NIST standards. If your tester is equipped with a capacitance measuring capability, then you will also need a Precision Capacitance Decade Box that is calibrated to NIST standards.
Note 5: When self-calibrating an HVX series model, the unit will not benefit from the services of 'Note 2' above.
The reliability of the equipment allows us to provide our customers with the best possible products. We've also found that the customer service and response time from CAMI Research to be excellent. All in all a company I would highly recommend using.
Eric Lutz, VP Production
Galaxy Wire and Cable, Inc.
Hear more from our customers ...
Yes, it is a non-standard calibration service option, and it may add up to one business day to your overall calibration service. If this is a requirement, be sure to mention it when you call for your calibration quotation and to include it on your PO when you book your service. It is imperative that your purchasing department email us this PO in advance of shipping the unit to us. See below for complete ordering details.
1 - Call CAMI Research at 978-266-2655 to arrange for return
of your unit. We need to confirm the name of the technical contact person
for CableEye, and your company's address and phone number.
(When shipping from an international location where we have a representative, please contact your local distributor first.)
2 - We will confirm the cost of the calibration, either by phone or by sending you a quote. Please provide us the serial number of the tester or testers that you want to calibrate, and be sure to mention whether As-Found data service is required.
3 - Your purchasing department must email us a PO in advance of shipping the unit to us. Your company pays transportation in both directions. We will accept VISA, MasterCard, and American Express if that is more convenient. Be sure to specify the method of return shipment you want us to use (Ground, 3rd day, 2nd Day, Overnight). Note: return shipment by FedEx Ground is not an available option, and return shipment options for International customers are FedEx or UPS, Priority or Economy.
4 - When you ship the system, review and circulate this preparation checklist to your packing/shipping department. [Remember to email us the tracking number.]
5 - We will generally complete the calibration and return the equipment to you within 1-2 business days of receiving it. For low voltage testers under warranty, we offer an expedite option for a fee:
+1 978 266 2655
CableEye is a highly versatile, expandable and upgradable diagnostic and Pass/Fail check Cable and Harness Test System that’s PC-based. It’s used for assembly, prototyping, production, and QC of standard or custom wire cables and harnesses The entire suite of products is powered by the same easy-to-use operating software and, with the help of its signature easy-to-interpret color-coded graphics, instantly identifies not only when there is a fault, but what type of fault and where. There are several models:
Low Voltage: M2U-basic, M2U
For diagnostic and Pass/Fail Testing - Find, display, log, & document: continuity (opens, shorts, miswires, intermittent connections).
Low Voltage: M3U, M3UH
For all of the above plus resistance (contact, isolation, embedded), and diodes (orientation, forward voltage, reverse breakdown).
Low Voltage: M4
For all of the above plus precision resistance (4-wire), and capacitance (twist wire relationship, length of cable, length to break, capacitors).
Low Voltage and High Voltage: HVX, HVX-21
For all as described for M3U, M3UH plus 4-Wire Kelvin Measurement and HiPot (dielectric withstand voltage and insulation resistance). Capacitance measurement (twist wire relationship, length of cable, length to break, capacitors) is an option.
Experience CableEye first-hand. See how your own cables and connectors can be auto-detected and accurately represented on our graphic-rich, touch screen compliant GUI. Find cable problems fast, and understand why customers tell us "... we can not live without CableEye" (Kabelservice), declaring it the "... best, easiest to use, system" (Digital Video Products).
Schedule your Free Trial now!
No credit card required. No commitment to purchase.